Conductivity measurement method
Automatic test(complete in 5 seconds)
Test Temperature Range
0.01%-1.99% ±0.1% 2.00%-3.00% ±0.2%
The maximum temperature of the measured object is 99℃
Test Completion Reminder
Display data stops flashing
Battery Replacement Reminder
Battery low voltage symbol flashes
2032 button battery
Low power consumption:Shut down automatically within 2 minutes
Temperature compensation range:0~60℃
Error range:0.01%-1.99% ±0.1% 2.00%-3.00% ±0.2%
How to use
1.Press the power button to turn on the meter and the short-line icon
2.Immerse the test object, the short- line icon stops flashing, and the test result is displayed.
3.The meter can test continuously for several times, to ensure the accuracy, it’s better to clean the meter before test.
4.After the test, please clean the meter with neutral detergent and clean water, wipe the moisture with soft cloth wet cloth,and put on the protective cove.
5.When testing, the head of the meter can not in contact with the container.
6.If the measurement is out of range, the instrument displays “HHHH”.
7.The instrument will automatic shut down without operation for 1 minute.
8.When testing pure water, the interface flash and no data display.