Low power consumption:Shut down automatically within 2 minutes
Temperature compensation range:0~60℃
Error range:0.01%-1.99% ±0.1% 2.00%-3.00% ±0.2%
How to use
1.Press the power button to turn on the meter and the short-line icon
2.Immerse the test object, the short- line icon stops flashing, and the test result is displayed.
3.The meter can test continuously for several times, to ensure the accuracy, it’s better to clean the meter before test.
4.After the test, please clean the meter with neutral detergent and clean water, wipe the moisture with soft cloth wet cloth,and put on the protective cove.
5.When testing, the head of the meter can not in contact with the container.
6.If the measurement is out of range, the instrument displays “HHHH”.
7.The instrument will automatic shut down without operation for 1 minute.
8.When testing pure water, the interface flash and no data display.
Calibration Method of Digital Salinity Meter
In the shut down state, insert the Salinity meter into the salinity calibration solution, press and hold the power-on key until the CCCC is present, and then return to the main interface after the automatic calibration is complete. The calibration is done
Tips: When testing pure water, the interface flash and no data display
Hotel Restaurant Family diet Food processing Desalination
Founded in 2008, Hangzhou Lohand Biological Technology Co., Ltd. Is a company that designs, develops, produces and sells water quality rapid test strips, reagents and precision testing instruments. It is located in Jianggan Science and Technology Park and relies on Zhejiang University Food Science and Technology Park and the West lake Science and Technology Park, focusing on technological innovation, has obtained a number of patents and produces provide customers with professional, fast, affordable and simple testing products and services to create a new history of water quality monitoring.
About Salinity meter
For rapid determination of salt (sodium chloride) solution weight percent concentration or refractive index. It is widely used in industrial sectors such as salt making, food, and beverage, as well as in agricultural production and scientific research.
Due to the simple operation, users can quickly read the salt concentration or seawater specific gravity. It can be used to manage the concentration of seawater used in oceans, fishing grounds, and farms, seawater or artificial seawater used in aquariums, and saltwater used to store fish.