Rapid Test Soil pH Meter 1. Introduction
Soil pH is an important factor in crop planting. Suitable pH in soil is the basic requirement for plant well-growing. Rapid test soil pH meter Soil pH-Moisture Meter can be used to test pH and moisture in soil. It’s portable, easy to use, and can be directly inserted into the soil.
Rapid Test Soil pH Meter 2. Technical parameter
rapid test soil pH meter
PH range: 3-8 pH, PH accuracy: ±0.2pPH, Resolution: 0.2pH
Moisture range: 10-80%,
Working condition: 5—50℃（41—122℉）
Rapid Test Soil pH Meter 3. Type list
The depth can be measured:6cm
Applicable to surface soil pH measurement
The depth can be measured:30cm
Applicable to the depth soil pH measurement
Rapid Test Soil pH Meter 4. Operation
1. Remove coverings such as stones, grass and leaves on the surface of soil. And remove the surface soil. If it’s too dry, add some water first. After 25~30 minutes, do measurement.
2. Before measuring, please use a piece of soft cloth to clean the metal surface of the probe. If the instrument is used at the first time, you’d better measure for several times, after that, read the value, which in order to avoid the influence on pH value and moisture value made by the protective oil layer of metal surface.
3. When measuring, insert the probe of the instrument into the soil. Pay attention, the probe should be completely inserted to the soil and make sure the probe and soil around contacted closely. In order to measure the soil pH value and moisture value accurately, the probe should be inserted in the soil for about 10 minutes. As nature of soil is different, the tightness of contact between probe and soil is also different. Thus, you’d better measure for several times and finally take the average value.
4. Press white button can measure the moisture range of the soil.
5. After using, please clean the probe.
Rapid Test Soil pH Meter 5. Attentions
(1). Before testing, scarify the soil, as hard soil will damage the probe.
(2). Do not keep the probe inserted in the soil for a long time (no more than an hour), otherwise the metal surface of the probe will be easily damaged.
(3). After using, please clean the probe.
(4). Keep the instrument away from objects with magnetism, and do not mix with other metal objects. Therefore, in order to avoid damage of the instrument.
(5). Can’t be used in liquid test, any liquid coming into the meter will damage the circuit.
(6). Don't use fingers to touch the probe, as grease on the finger may affect the conductivity of the probe between electrodes, which may lead to inaccurate measurement data.
(7). If the probe is loose, tight it by rotating clockwisely.
Rapid Test Soil pH Meter 6. Accessory
Rapid Test Soil pH Meter Our company:
Rapid Test Soil pH Meter Our service:
Rapid Test Soil pH Meter Certifications:
Rapid Test Soil pH Meter FAQ:
1.What about the transportation?
It depends on your requirement.We can delivery by air to your airport ,or by sea to your seaport ,and we can also delivery to your office address by express courier,and we also accept to co-operate with your ship agent or forwarder.
2.What after-sales service can be provided by Lohand ?
The after-sales service of Lohand is divided into two parts: technical support and maintenance.
3.What is the lead time for products order?
Usually, if small order, 2-3days after getting payment;if big order, it needs 10-20days after payment, it depends upon your quantity.