Turbidity measurement device WGZ Series dispersive optical turbidity gauges are used to measure the degree of light scattering from insoluble particulate matter suspended in water or transparent liquids, and to characterize the problem of suspended grain content in these substances. Turbidity measurement device can be widely used in power plants, water plants, domestic sewage treatment stations, beverage plants, environmental protection departments, industrial water, alcohol, pharmaceutical industry, turbidity measurement, popular prevention departments, hospitals and so on.
turbidity measurement device
Minimum Readout ( NTU)
Bench top type
1, zero turbidity water poured into the sample bottle to the scale line, then screw on cap, and clean the bottle water trace and fingerprints, it should be noted that the open not available when the hand directly with the bottle body, so as not to leave fingerprints, affect the accuracy of measurement.
2, will be packed with zero turbidity water sample bottle, into the sample holder, and to ensure that the specimen bottle scale line with quasi specimen holder white line of position, and then covered with a shading cover.
3, wait for stable after adjusting zero adjusting knob, so that the display is zero.
4, the same device calibration using 100NTU standard solution, and is placed in the sample holder, adjustment knob, so that the display as the standard value 100.
5, 2, 3, 4, repeat steps, to ensure zero and calibration values are correct and reliable.
6, into the sample bottle, and stable to note the water turbidity value.
1,after starting up, display
A、power line and socket connection is bad or loose
A、fastening socket or for power line
A, light damage
B, internal connector loose
C, electrical system fault
3,measurement instability or drift
A, solution with bubbles or particles in does not stop moving
B, circuitry within the apparatus of damp
C, the outer surface of the bottle water sample
D, external interference
E, the supply voltage instability of
A, re-sampling or extended reading time
B, prolonging start preheat time preheating dehumidification
C, dry specimen bottle
D, eliminating the interference source
E, exclusion of unstable factors
4.zero, less than zero
A, adjustable zero without the use of zero turbidity water
B, zero potentiometer damage
C, zero range migration
D, light path offset
A, should adopt zero turbidity water
B, replacement potentiometer
C, W2 potentiometer adjustment board
5,not to the correction value
A, standard solution standard value is not accurate
B, correction potentiometer is damaged
C, range migration correction
D, light path offset
A, accurate preparation of standard solution
C, W1 potentiometer adjustment board
A, due to the transport screen vibration caused by poor contact
B, LCD LCD screen damage
C, circuit board fault
1.What delivery / shipping way do you use?
For samples or trial small order, we can send them to you by Air Express, such as DHL, FEDEX, UPS, TNT or EMS;
For large amount goods, we can book the ship to your sea-port by sea-shipping;
2.What payment terms you accept?
EXW or Price is available, and Paypal/TT/Western union is workable.
3.How long can we receive the response after sending inquiry?
we are online anytime, you can contact us to know within 24 hours.
4.What after-sales service can be provided by Lohand ?
The after-sales service of Lohand is divided into two parts: technical support and maintenance.
5.What should I do if some parts broken?
Dear customer, please do not worry, we have 12months warranty except wear parts. You can also buy parts from us after 12months.